1). The flow sensor can be installed indoors or outdoors to avoid high-voltage lines, rotating mechanical equipment, toxic and harmful environments, strong mechanical vibrations and other environments that endanger the safety of people and instruments, as well as temperature and humidity environments that are inconsistent with the use conditions of the flow sensor. Choose a place that is safe, easy to install, debug and overhaul, and has a better environment. (2). Install the flow sensor in the design 1). The flow sensor can be installed indoors or outdoors, avoiding high-voltage lines, rotating mechanical equipment, toxic and harmful environments, strong mechanical vibrations and other environments that endanger the safety of people and instruments, and use it with the flow sensor Temperature and humidity environment that does not match the conditions. Choose a place that is safe, easy to install, debug and overhaul, and has a better environment. (2). When designing the installation location of the flow sensor, in addition to considering the conditions of the straight pipe section, there should be enough operating space around the flow sensor to facilitate installation, debugging and maintenance. In particular, it is necessary to leave a safe and convenient operating space for people. When the instrument is installed at high altitude, an operating platform should be made to ensure the safety of people operating at high altitude. (3) The flow sensor can be installed on horizontal, vertical or inclined pipelines. But when measuring liquid, the flow sensor installed on a vertical or inclined pipeline must ensure the flow direction of the liquid from bottom to top to ensure that the pipeline is filled with liquid and offset the impact of additional gravity (see Figure 9). (4). Long straight pipe sections should be left upstream and downstream of the flow sensor. The size of the straight pipe section varies according to the upstream pipe type. According to the JB/T9249-1999 vortex flow sensor standard, Table (3) lists the minimum straight pipe length requirements for different pipe types upstream of the sensor. (5). The inner diameter of the pipe where the flow sensor is installed must be the same as the inner diameter of the flow sensor. Otherwise, the upstream and downstream pipelines where the flow sensor is installed must be reduced in diameter. The principle of selecting the pipe diameter is: the pipe inner diameter must be consistent with the nominal inner diameter of the flow sensor. The relative error of the two inner diameters should be controlled within ±1%. On the premise of ensuring the pressure strength, the nominal pressure of the straight pipe diameter within 4Mpa of the LU vortex flowmeter, LUGB vortex flow sensor and the instruction manual refers to the table (4). (6). It will not affect continuous production when it is disassembled for maintenance. Under conditions, the installation method with bypass pipe is recommended. See Figure 12. (7). No regulating valve or half-open valve should be installed upstream of the flow sensor, so as not to affect the accuracy and stability of the flow sensor. The regulating valve or half-open valve should be installed outside the 5DN downstream of the flow sensor. See Table 3 and Figure 12. 'DN' is the nominal diameter of the flow sensor
- E-mail: firstname.lastname@example.org - Wechat : +86 13821000231
- Whatsapp: +86 13821000231
- Factory address: No.12 outer ring industry park zhong bei industry park (northern area), xiqing district,tianjin.